Indications: The PureWick™ Urine Collection System is to be used with the PureWick™ External Catheters which are intended for non-invasive urine output management.
Contraindications: Do not use the PureWick™ Urine Collection System with PureWick™ Female External Catheters on individuals with urinary retention.
Warnings: Do not use the PureWick™ Female External Catheter with bedpan or any material that does not allow for sufficient airflow. To avoid potential skin injury, never push or pull the PureWick™ Female External Catheter against the skin during placement or removal. Never insert the PureWick™ Female External Catheter into vagina, anal canal, or other body cavities. Discontinue use if an allergic reaction occurs. After use, this product may be a potential biohazard. Dispose of in accordance with applicable local, state and federal laws and regulations.
Precautions: Not recommended for patients who are: Agitated, combative, or uncooperative and might remove the PureWick™ Female External Catheter. Having frequent episodes of bowel incontinence without a fecal management system in place. Experiencing skin irritation or breakdown at the site. Experiencing moderate/heavy menstruation and cannot use a tampon. Do not use barrier cream on the perineum when using the PureWick™ Female External Catheter. Barrier cream may impede suction. Proceed with caution in patients who have undergone recent surgery of the external urogenital tract. Always assess skin for compromise and perform perineal care prior to placement of a newPureWick™ Female External Catheter. Maintain suction until the PureWick™ Female External Catheter is fully removed from the patient to avoid urine backflow.
Safety and Warnings: Always unplug PureWick™ Urine Collection System before cleaning or when not in use. Do not immerse the PureWick™ Urine Collection System in water. As with most electrical devices, electrical parts in this system are electrically live even when the power is off. To reduce the risk of electric shock, if the PureWick™ Urine Collection System falls into water, unplug immediately. Do not reach into the water to retrieve it. Contains small parts that may cause choking. Keep out of reach of children. Keep cords and tubing out of the reach of children to avoid the risk of strangulation. Discontinue use if an allergic reaction occurs. Not recommended for users who are experiencing skin irritation or skin breakdown in device contact areas. WARNING: This device should not be used in oxygen rich environments or in conjunction with flammable anesthetics. If the PureWick™ Urine Collection System is dropped or tipped over spilling urine, unplug the unit and carefully inspect for loose or damaged parts before resuming use. Contact Customer Support at 1-888-201-1586 if any damage is observed. The pump tubing connecting the PureWick™ Urine Collection System to the collection canister may experience light condensation inside the tube. This is not unusual and does not affect function. However, if urine or water has streamed into the pump, discontinue use and contact Customer Support at 1-888-201-1586. Although the canister can hold up to 2000cc (mL), the urine should be emptied regularly from the collection canister before volume reaches 1800cc (mL). Failure to empty canister before urine overflow may cause damage to the PureWick™ Urine Collection System and is not covered under the warranty. It is important that the port connections be connected correctly for proper operation of the PureWick™ Urine Collection System. Use of this equipment next to or stacked with other equipment should be avoided because it could result in improper operation. If such use is necessary, this equipment and the other equipment should be observed to verify that they are operating normally. Portable radio frequency (RF) communications equipment (including peripherals such as antenna cables and external antennas) should be used no closer than 12 inches (30cm) to any part of the PureWick™ Urine Collection System, including cables specified by the manufacturer. Otherwise, degradation of the performance of this equipment could result. Use only PureWick™ Urine Collection System accessories with this device. Incompatible parts or accessories can result in degraded performance and will void the warranty. Do not use accessories past expiration date indicated on package labeling. Use only the PureWick™ Urine Collection System A/C power cord with the device. Use of an alternate consumer style A/C power adapter or an extension cord may cause damage to device and electrical shock or injury and will void the warranty. Do not place PureWick™ Urine Collection System or its cord across walkways creating a tripping hazard. For PW200: The battery cells used in this device may present a fire or chemical hazard. To minimize the risk of damaging the battery inside the PureWick™ Urine Collection System, do not use the device outside the indicated operating temperature range of 41°F – 104°F (5°C – 40°C). The battery is not intended to be replaced; replacement could result in a hazard. To reduce the risk of electrical shock or injury, do not disassemble this unit. No modification of this equipment is allowed.
When you visit any website, it may store or retrieve information on your browser, mostly in the form of cookies. This information might be about you, your preferences or your device and is mostly used to make the site work as you expect it to. The information does not usually directly identify you, but it can give you a more personalized web experience. Because we respect your right to privacy, you can choose not to allow some types of cookies. Click on the different category headings to find out more and change our default settings. However, blocking some types of cookies may impact your experience of the site and the services we are able to offer. Cookies Policy
Necessary cookies are absolutely essential for the website to function properly. This category only includes cookies that ensures basic functionalities and security features of the website. These cookies do not store any personal information.
Any cookies that may not be particularly necessary for the website to function and is used specifically to collect user personal data via analytics, ads, other embedded contents are termed as non-necessary cookies. It is mandatory to procure user consent prior to running these cookies on your website.